- Poster presentation
- Open Access
P101: Product dose considerations for real-world hand sanitiser efficacy
© Hines et al; licensee BioMed Central Ltd. 2013
- Published: 20 June 2013
- Healthcare Worker
- Hand Hygiene
- Foam Format
- Alcohol Base Hand
- Hand Sanitiser
Alcohol based hand rubs (ABHRs) are extremely effective at reducing microbial contamination and are central to established best practices for hand hygiene. Modern dispensing systems have brought benefits such as hygienically sealed cartridges with integral pumps for dosing liquids, gels or foams. A remaining issue concerns the measured efficacy of such products and their use in practice as dictated by pump volume.
ABHRs for professional use must demonstrate efficacy through standard in-vivo tests; either EN1500 or ASTM E1174. Products typically pass such tests using standard doses not necessarily related to real use by healthcare workers. In this study we set out to identify the optimal dose for an ABHR product based on observation of real behavior.
Data from the DebMed GMS Hand Hygiene Monitoring System was used to establish product dose applied in clinical settings via the number of dispenser presses per hand hygiene event. In a separate study healthcare workers applied ABHR using best-practice methodology to establish product drying time as a function of dose. Hand coverage was assessed using a laboratory method based on product coverage. In-vivo efficacy was established using both ASTM E1174 and EN 1500 methods.
Healthcare workers intuitively calibrate ABHR dose based on drying time, hand coverage and product ergonomics. In our studies using Deb dispensers, over 90% of healthcare workers used a single pump of ABHR, even when multiple pumps are indicated. Our studies established that 1.5ml of ABHR in foam format dries in approximately 30 seconds and fully covers most hands. Both ASTM 1174 and EN 1500 tests confirmed that 1.5ml of Deb ABHR in foam format with 30 seconds contact time was sufficient to provide effective hand hygiene.
Product drying time, hand coverage and measured efficacy data are considered to determine the optimal dose for an ABHR in foam format dispensed from a sealed cartridge wall-mounted unit. This is combined with observed behaviour of healthcare workers to design a pump that delivers the optimal dose in a single press, connecting for the first time measured in-vivo efficacy standards with real-world use to ensure effective sanitization.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.