- Poster presentation
- Open Access
P295: Endoscopes’ maintenance: results from an audit conducted in Beni-Messous University Hospital Algiers in 2012
© Brahimi et al; licensee BioMed Central Ltd. 2013
Published: 20 June 2013
Medical devices used in endoscopy should be disinfected according to best practices to eliminate microorganisms on their surfaces.
Evaluate the practical disinfection of endoscopes.
The audit was conducted from 03 to 19 January 2012, in the services using endoscopy. Data collection was done through observation and interviews of staff responsible for the disinfection of endoscopes from a predetermined rubric. Entry and data analysis were performed on Epi-info6.
7 services use Endoscopy (Otorhinolaryngology, Pulmonary and Allergy Medicine, Allergy, General surgery, Pulmonary function tests, CPPA, Internal Medicine, Pediatric). 15 people responsible for the disinfection of endoscopes were audited. 66.7% are nursinggraduates, 21.4% are instrumentalists. More than 50% have an average experience of 14 ± 9 years. Disinfection of endoscopes every morning before first use is achieved in 40% of cases. The seal is tested in 74% of cases.
The second cleaning is never done. The final rinsing takes place with tapwater, drying with a sterile field in 40% and by blowing compressed air in 20% of cases.
The storage isn’t daily cleaned and disinfected. Periodic maintenance of the unit is done neither by the manufacturer nor the distributor.
This audit revealed the lack of necessary materials, and non-compliance with the disinfection protocol of endoscopes in some services and the absence of microbiological testing.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.