- Open Access
Efficacy of disinfectant-impregnated wipes used for surface disinfection in hospitals: a review
Antimicrobial Resistance & Infection Controlvolume 8, Article number: 139 (2019)
“Ready-to-use” disinfecting wipes (also known as pre-impregnated disinfecting wipe) are broadly used in food industry and domestic situations. Their application in hospitals and healthcare centres for decontamination of medical devices and surfaces is steadily increasing because of their convenient implementation in practice and reliable performance. Beside their acceptable compliance and easy application, literature reported the disinfection failure due to the interaction between textile substrate and active ingredients, which can highly increase the risk of an infection outbreak. This review aims to call attention to the wide range of variables affecting the disinfectant-impregnated wipes’ (DIWs) disinfection performances in hospitals.
A systematic literature search based on the five categories i. wipes, ii. disinfectants, iii. Application methods, iv. interaction between wipes and active ingredients and v. wiping strategy which can possibly influence the disinfection effectiveness of DIWs was conducted by Google scholar. Studies regarding the efficacy evaluation of DIWs in clinical applications were also reviewed from the National Centre for Biotechnology Information database.
Variables that impact on the disinfection performance of disinfectant-impregnated wipes in surface disinfection in hospitals were summarised and critically discussed. In addition to the information, current disinfectant-impregnated wipes’ decontamination efficacy test standards were reviewed, and different standards exposed some disadvantage in their testing design.
Various parameters contribute to the impact of DIWs disinfection performance in practice. The interaction between disinfectant active ingredients and the wiping materials barricades their broad application in hospitals. More studies of the DIWs’ disinfection efficacy in clinical practice are in need. Current standards evaluating the DIWs’ efficacy are required to improve for more realistic condition simulation and differentiating between mechanical removal of inoculum from a surface and chemical inactivation of the test microbe.
Healthcare-associated infections (HCAIs) caused by the transfer of nosocomial pathogens from high-touch environmental surfaces and medical devices are responsible for significant patient morbidity, mortality and economic cost [1,2,3,4,5]. More recently, evidence shows nosocomial pathogens, including methicillin-resistant Staphylococcus aureus (MRSA), norovirus, Clostridium difficile, vancomycin-resistant Enterococcus, and Acinetobacter species etc. shed by patients can contaminate hospital surfaces at concentrations sufficient for transmission, surviving for extended periods and persisting despite attempts to remove them [5,6,7,8,9,10,11].
An effective cleaning and disinfection practice, such as chemical disinfection, heat, and ultraviolet germicidal irradiation etc. play a key role in preventing cross-contamination [12, 13] and spread of HCAIs [14,15,16,17,18,19]. Among all the surface disinfection approaches, the utilization of chemical disinfectant is broadly diffused in food industry, hospitals and healthcare centres because of its easy application and broad spectrum of microbicidal activity [20, 21]. In the application of disinfectant in practice, the “ready-to-use” disinfecting wipes (RTUDW) (also reported as pre-impregnated disinfecting wipes, pre-saturated towelette and pre-wetted disinfecting wipe in some literatures) are increasingly accepted for decontamination of high-touch surfaces because of their convenient implementation and reliable performance [22,23,24].
Although disinfecting wipes have been widely used and spread in hospitals for decontamination of medical devices or high-touch environmental surfaces , the effectiveness of their disinfection performance is always in discussion.
Disinfectant-impregnated wipes (DIWs) basically consists of towels saturated with diluted disinfectant as well as other chemical products such as surfactants, preservatives, enzymes, and perfumes etc. When two materials encounter, the interaction between each other is not negligible and often has an influence on their original function. The factors that could possibly influence the system’s disinfection efficacy focus on the following aspects: wipe, disinfectant, application method, the interaction between each other as well as the wiping strategy and storage time. Going through the studies regarding the efficacy of DIWs in clinical practice, the authors believe that more attention needs to be addressed in this topic. The review also points out the need for improvement for disinfecting wipe decontamination efficacy testing standards. Since it is an important validation step before the disinfectant-impregnated wipe products launched into the market and further used in the hospital.
Several parameters influencing the antimicrobial efficacy embracing the external factors for instance target surface (material, organic load), target microorganism, ambient environment (temperature, humidity), and internal factors such as the disinfectant (type, concentration), wipe (material type, construction/fibre architecture), application method, wiping strategy, are investigated by numerous researchers. A systematic literature search focusing on the internal factors were conducted by Google scholar. General efficacy study of DIWs in clinical practice was searched on NCBI (National Center for Biotechnology Information) database. For additional information related to the efficacy testing protocols, standards were explored and reviewed under the scope of EU standards issued by the European Committee for Normalization (CEN), Technical Committee 216 (TC 216) and US standards by the United States Environmental Protection Agency (EPA) Office of Pesticide Programs (OPP) in cooperation with AOAC International and ASTM International. In addition, guidelines i.e. Guidance on the Biocidal Products Regulation from European Chemicals Agency (ECHA) and Guideline for Disinfection and Sterilization in Healthcare Facilities by Healthcare Infection Control Practices Advisory Committee (HICPAC) of Centres for Disease Control and Prevention of USA (CDC) are also included for the review.
Data were extracted by two authors based on the examination of the titles and abstracts obtained from Google Scholar and PubMed database. Meanwhile, one author reviewed the abstracts to check if the material fulfils the criteria of the work. Subsequently, full articles deemed necessary for the review were obtained and reviewed by another author.
Results and discussion
The disinfection process of a disinfectant-impregnated wipe (DIW) can be divided into two parts that constitute the overall decontamination activity. One part is related to the microorganisms taken away by the wipe itself by means of mechanical action. The other part is related to the active microbicidal action of the disinfectant solution released by the wipe on the surface. The parameters that influence its efficacy, as well as the effectiveness studies of DIWs in literature are exhibited as follows.
The wipe for disinfection is mostly made of textile materials, including, but are not limited to, cellulosic fibres (cotton, woodpulp, viscose, lyocell) and thermoplastic fibres (polyethylene terephthalate, and polypropylene). Particularly for disposable wipes, the raw materials are normally inexpensive like cellulosic fibres and polyolefin fibres. Cellulosic fibres are used to ensure high water retention and storage capacities and polyolefin fibres are accountable for high tensile strength, abrasion and solvent resistance . The majority of wipes for surface disinfection in the market are made of blends of polyester and viscose fibres/woodpulp [26, 27].
Wipes offer a cleaning procedure by the mechanical action of wiping, which can remove the organic debris along with the disinfection activity.
Likewise, the microorganism can be mechanically removed by the wipe. However, attention should be paid to the transfer of microorganism to other parts of the surface. The removal of the microorganism depends on the inherent properties of the wiping material such as surface energy, fabric structure and fibre types as well as by the applied pressure force, the geometry of the mechanical action, the number of passages and type of microorganism adhesion mechanism [28, 29]. As stated before, it is also important to consider that during the wiping action some microorganisms could be just transferred in another place of the treated surface instead of being removed. This transfer depends by the wipe retaining ability and by the bactericidal activity of the disinfectant adsorbed into the wipe .
The disinfectant solution released by the wipe on the target surface is mainly responsible for the bactericidal activity. The quantity and concentration of active ingredient and the amount of the solution remaining on the surface are important efficacy indicators and depending on the interaction between the wipe and disinfectant. Also, the amount of released solution is highly dependent on the wipe absorbent property. Without any doubt, wipe plays an important role in decontamination of the target surface. Table 1 below looks at available advance wipes.
Disinfectant as the main constituent for disinfection action has a crucial impact on the decontamination process. The antimicrobial activity of disinfectants performs in two different ways: growth inhibition (e.g. bacteriostatic, fungistatic) and lethal action (sporicidal, bactericidal, fungicidal, and virucidal effects) . Disinfectants comprise a wide variety of active chemical agents (biocides). The active ingredients found in the market are generally alcohols, chlorine, aldehydes, peroxygens, and quaternary ammonium compounds . Every type of disinfectant presents some advantages and disadvantages allowing, or not, its use in wipes.
Alcohol is cheap and easy to obtain allowing an efficient wetting of the surfaces with a rapid bactericidal effect without bacteriostatic action and relevant toxicity issues. However, it is highly inflammable, corrosive to metals with lack of efficacy in the presence of organic debris and tends to swell and harden rubber and certain types of plastics. It is not sporicidal and has low effectiveness in the inactivation of some type of virus. Moreover, due to its high volatility, it is difficult to ensure enough contact time in open systems [38,39,40,41,42].
Under the chemical family of chlorine compounds, the most used disinfectants are hypochlorite, chlorine dioxide and the chloramine-t trihydrate. Hypochlorite is the most used chlorine disinfectants especially because of its low cost and fast mode of action. It displays a large bactericidal spectrum with no toxic residues, and it is not affected by water hardness. However, it is also corrosive to metals (> 500 ppm), easily inactivated by organic matter, irritating and burning for skin, eyes and mucous membranes. It can discolour and bleach textiles and can become very dangerous in contact with ammonia or acid due to the generation of toxic chlorine gas. Chlorine dioxide also shows a wide spectrum of biocidal activity including mycobacteria with short contacts time. It provides prolonged bactericidal effect than chlorine due to its high retain of antimicrobial active ingredients but with long-term use can damage the outer plastic coat of some insertion tubes. Chloramine-t trihydrate it is able to retain chlorine for a long time which results in a more prolonged bactericidal effect, however, occupational asthma has been reported due to its prolonged exposition [25, 43,44,45].
Hydrogen peroxide presents a satisfying germicidal activity including bacterial spores (with longer contact time). It is relatively environmentally friendly due to its fast degradation. An accelerated hydrogen peroxide (AHP) was specifically developed for widened material compatibility and application variability. It can cause chemical irritation resembling pseudomembranous colitis. Peracetic acid (PAA) has a rapid action against all microorganisms (inclusive spores at low temperatures) at low concentration. It is very efficient even in the presence of organic matter, without producing residues. However, it is unstable, particularly when diluted, and is corrosive to copper, brass, bronze, plain steel, and galvanized iron (corrosion decline by additives and pH modifications) [46,47,48,49,50,51].
Quaternary Ammonium Compounds (QACs) are the most commonly used disinfectant in ordinary environmental surfaces with a good cleaning and deodorization property and most important a broad spectrum of biocidal and sporostatic activity (lipid, enveloped viruses). Incorporation of quaternary ammonium moieties into polymers showed an effective antimicrobial effect against biofilm. Nowadays, QACs are the most used disinfectant in wipes. However, they also have some drawbacks such as a susceptibility to high water hardness and low efficacy against gram-negative bacteria and non-enveloped viruses. Moreover, numerous studies showed that the adsorption of QACs onto the cotton substrate wiping material could lead to the failure of the disinfection process [47, 52,53,54,55,56,57,58].
Summarise the information, the preferred disinfectants used for disinfectant -impregnated wipes in the market are quats-alcohol wipes, hydrogen peroxide wipes, hypochlorite wipes due to their user-friendly feature. Emphasis is drawn to the safety issue for the staff during their application. These DIWs are extensively approved to be employed in hospitals and healthcare centres. More information regarding the disinfectant and their application evaluation can be found in Table 2.
When applying the surface disinfectant on the target surface, the approaches can be generally divided into two groups: i) without mechanical action, e.g. total immersion and directly spraying, and ii) with mechanical action, e.g. spray & wiping, dipping & wiping, and soaking & wiping . The main benefit from the mechanical action is its ability to remove the organic debris that could hinder the disinfection action and DIWs falls in the category with mechanical action.
The “Spray and Wipe” method starts with a direct spray of the disinfectant solution with an aerosol or trigger sprayer on the target surface, followed by a wipe of the target surface. The spray action allows direct contact of the disinfectant solution with the target surface. However, there are several drawbacks such as possible overspray, difficulty in covering surfaces (undersides of bedrails), and generation of atomized disinfectant in the air that can subsequently be breathed by workers and patients . Due to the flammability of numerous sprayed disinfectants, the presence of open fires during use have to be taken into account .
“Dip and Wipe” means dipping a dry towelette into one disinfectant solution for 5–10 s, wring out the excess solution and directly use it for disinfecting hard surfaces. The short contact time that the wipe spends in the disinfectant solution can limit the concentration of active ingredients applied on the target surface. A towelette carrying an insufficient amount of surface disinfectant may lose its antimicrobial activity and later becomes itself a potential vehicle of pathogen transmission . In addition, the inappropriate reuse of the towelette may promote the accumulation of microorganisms and raise the risk of cross-contamination during the disinfection process [63,64,65].
“Soak and Wipe” method, also known as “bucket method”, was widely used for disinfection processes in hospitals. Similar to the “Dip and Wipe” method [66, 67], the towelette is soaked into disinfectant solution from 10 min up to 8 h instead. Before use wring out the excess solution and directly applied to a hard surface. The “Soak and Wipe” method was the most prevailing methods among all, owing to its acceptable performance and easy application. It allows a relatively long contact time ensuring enough active ingredient load in the towelette before use. Nevertheless, there are some studies reporting possible interactions between wipes and disinfectant due to the longer soaking time resulting in reduced antimicrobial activity of disinfectants [52, 68, 69]. Moreover, a chemical binding of the disinfectant to the wipe could lead to a decrease of disinfectant concentration in the bulk solution . As indicated in the aforementioned method, improper reuse of the towelette can result in cross-transmission of pathogens on the treated surfaces [63,64,65].
Ready-to-use disinfecting wipe (Abbreviated as RTUDW) is a pre-wetted towelette containing disinfectants, antiseptics, surfactants etc. in a sealed package ready for use in surface disinfection up to 1 month (shelf life can be longer). This method is also known as “pop up” wipe in hospitals. The use of RTUDW is steady increasing partially profits from the rapid development in nonwoven technologies, which provides a relatively good cost performance . The RTUDW is designed to be used without any preparation time. Considering the compliance, employee time, and costs, RTUDW is highly recommended for the surface disinfection . It has been tested in many research projects to be proved to possess good antimicrobial effect in several conditions [40, 64, 72]. RTUDW is disposable, which eliminates the possible contaminations and transfer of pathogen due to towelettes reuse . However, the longer storage time could increase the probability of losing antimicrobial activity due to the possible binding of active ingredients onto the towelettes or by the degradation of the active ingredient . Some research shows RTUDW’s bactericidal efficacy decreases on wiped larger surface areas . In addition, the disposable property could be a problem with waste management.
All above-mentioned application methods can be found in use in practice for different surfaces. “Spray and Wipe” and “Dip and Wipe” are not recommended for surface disinfection in general due to many drawbacks previously listed. “Soak and Wipe” is still commonly used in the hospital for daily cleaning and disinfection of large high-touch environmental surfaces such as floors, tables, lockers, examination couches. However, in this method, a potential interaction between disinfectant solution and wiping material can decrease the efficacy. The most prominent method is ready-to-use disinfecting wipes. Considering the antimicrobial efficacy of commercial wipes is already qualified by required standards before released into the market, there is less possibility of disinfection failure with this method. Nevertheless, ageing of the products needs to be further investigated as well as other parameters (e.g. wiping area, wiping passage, etc.) during the wiping process should be clarified by the manufacturer on the package.
Interaction between wiping material and disinfectant
A few investigations have been performed evaluating the interaction between wiping material and surface disinfectant. Unfortunately, nearly all of them were exclusively focusing on the interaction between quaternary ammonium salts (quats) and cotton substrate. Bloss et al. (2010) have classified the absorption of active ingredients onto textile substrate by testing three different surface disinfectants and four different types of fabrics. They found out that the exposure of diluted surface disinfectants to various types of fabrics resulted in considerable adsorption of active ingredients . Additionally, Boyce et al. (2015) found that several factors, including the soaking time and quats binding to specific wiping material, influence the efficacy of quats-based disinfectants. However, their experimental design showed two severe limitations: i) the wipes were taken out for quats concentration test in chronological order and the adsorption of wipes accounting for the decrease of quats concentration in the bucket was not taken into consideration; and ii) the lack of microbiological tests can hardly determine whether the low concentrations of quats released from the three wiping materials resulted in less potent reduction of bacterial counts on surfaces . The investigation of Hinchliffe et al. (2016) may be the first comprehensive study of the possible parameters from both perspectives of textile substrate and disinfectant solution influencing the quats binding degree onto a cotton substrate. They found that the amount of alkyl-dimethyl-benzyl-ammonium chloride (ADBAC) depleted from solution varied with the liquor ratio, pH, temperature, concentration of electrolytes and type of pre-treatment applied to the textile substrate. However, their investigation only measured the adsorption of active ingredients onto textile substrate in the bulk solution instead of the loss of active ingredients during the application stage (resulting from the binding of the active ingredients on the textile substrate). [52, 75]. Later, they demonstrate that quats adsorption onto cotton substrate can be minimized and maintain the efficacy against gram-negative (Pseudomonas aeruginosa) and gram-positive (Staphylococcus aureus) bacteria. 
In summary, disinfectant concentration, material compatibility, contact time, liquor ratio (wipe mass/disinfectant solution volume), an additive of other chemistries, and temperature are possible parameters impacting on the interaction of disinfectant and wipes.
Several studies have shown that high-touched surfaces and devices can serve as a route for transmission of pathogens [77,78,79]. However, proper disinfection protocols and wiping strategies are still in development.
Wiping strategy includes the applied pressure force, wiped surface area, the geometry of the mechanical action, the number of passages etc. One recent study from A. M. West et al. tested the bactericidal efficacy of ten ready-to-use disinfectants in the form of pre-wetted towelettes . The objective of the study focusses on the impact of surface area(s) wiped on its bactericidal efficacy. The result implicates a larger wiping surface area may lead to decreased bactericidal efficacy. However, rare attention is given for this factor, especially a severe lack of consideration from the efficacy testing standards. Detailed information is given in the next section.
Standards for disinfecting wipes’ efficacy test
In the last decades, numerous regulations and standards have been issued by various organisations for testing the efficacy of the disinfectants. The standards cover the most important factors that influence the effectiveness of a disinfectant, such as the target microorganism (bactericides, mycobactericides, sporicides/sterilants, fungicides, tuberculocides and virucides), the target surface (tile, stainless steel, wall panels, glass, etc.) and the application strategy (liquid, with wipe or spray method). Many protocols have been designed to validate the disinfectant’s efficacy at the concentration commonly used against a panel of clinically significant microorganisms on the surfaces most routinely disinfected.
In EU, the disinfectant efficacy test is regulated and issued by the European Committee for Normalization (CEN), Technical Committee 216 (TC 216) under the work program “Chemical Disinfectants and Antiseptics” . Two phases were developed for assessing the disinfectant effect: 1) Phase 1 is mainly suspension-based tests for the basic evaluation of disinfectant efficacy against different microorganisms, apart from mycobacteria, under clean conditions. It is applied to evaluate the bactericidal (EN 1040), sporicidal (EN 14347) and fungicidal (EN 1275) activity of chemical antiseptics and disinfectants when appropriate standards are not available. It is a minimum requirement for the assessment of basic biocidal activity under generic conditions (food, industrial, domestic and institutional, medical and veterinary areas). 2) Phase 2 is designed for evaluation of the bactericidal, sporicidal, fungicidal and virucidal, activity of chemical disinfectants applied individually in specific conditions such as food, industrial, domestic, institutional, medical and veterinary areas. In the scope of Phase 2, European Norm is divided into two steps. Step 1 is a suspension test while step 2 is a carrier-based test. Both as suspension-based tests, Phase 2, step 1 test is prior than Phase 1 not only because the application area is more specific in the test but also because it introduces the dirty conditions in testing the performance of surface disinfectant with the involvement of organic debris (Phase 1 only tests clean conditions). The dirty condition is able to demonstrate if a product (surface disinfectant) reacts with other substances such as proteins. Unfortunately, the suspension-based test is far away from the disinfectant performance in real practice. Consequently, carrier-based tests were developed to fulfil the need for disinfectant efficacy evaluation to various surfaces (instruments, surfaces, etc.) under practice-oriented conditions. Notably, in the carrier-based test, there are standards used for non-porous surfaces and porous surfaces with and without mechanical action. In the case of disinfecting-impregnated wipes, it applies to the standards for non-porous surfaces with mechanical action. In conclusion, standard EN 16615 is the most suitable one.
In the US, the United States Environmental Protection Agency (EPA) Office of Pesticide Programs (OPP) has the responsibility for regulating antimicrobial products used for treat and decontamination inanimate surfaces. In 1998, they first published the Product Performance Test Guidelines, OPPTS 810.2100 Products for Use on Hard Surface – Basic Efficacy Data Requirements used for efficacy testing of disinfectants in collaboration with AOAC International. Lately, it is amended as OSCPP 810.2200: Disinfectants for Use on Hard Surfaces – Efficacy Data Recommendations in September 2012. EPA recommended the carrier tests and use-dilution tests for assessment of disinfectant effectiveness for medical use surface disinfection . Up to date, in cooperation with AOAC International and ASTM International, antimicrobial testing methods & procedures are well documented and specified from EPA’s microbiology laboratory for antimicrobial formulations in the form of liquid, spray and towelette, against Staphylococcus aureus, Pseudomonas aeruginosa, Salmonella choleraesuis, Mycobacterium bovis (BCG), Clostridium difficile, Trichophyton mentagrophytes, non-enveloped viruses (i.e. parvovirus, noroviruses) and Pseudomonas aeruginosa or Staphylococcus aureus biofilm. Detailed testing methods are discussed in the following paragraphs.
AOAC Use Dilution Test is a standard operating procedure requested by EPA for evaluating liquid and dilutable liquid disinfectants for hard surfaces. Different series were developed for different microorganism tests - 955.14 (Salmonella enterica), 955.15 (Staphylococcus aureus), and 964.02 (Pseudomonas aeruginosa). The AOAC Use-Dilution Test is a relatively facile method to operate. However, it cannot demonstrate the use of disinfectants in practical conditions. Other methods are also specified by US EPA such as the AOAC METHOD 965.12 Tuberculocidal Activity of Disinfectants, which is a modified version of AOAC Use- Dilution test method applied to justify tuberculocidal efficacy claims for disinfectants. Due to the slow growth rate of the test microorganism (60 days’ incubation time plus an additional 30 days), the test is susceptible to contaminations. AOAC METHOD 955.17 Fungicidal Activity Method, which is designed to access the effectiveness of the disinfectant’s fungicidal activity. In the test, the highest acceptable dilution to disinfect a surface that is contaminated with the fungi in the given contact time is determined. AOAC METHOD 966.04 Sporicidal Activity of Disinfectants is developed to substantiate the sporicidal efficacy of high-level disinfectant or sterilant. By enumerating the number of spores per carrier and the drying the spores on the surface, the test is recognised as a more robust challenge for the rigour of the disinfectant. AOAC METHOD 960.09 Germicidal and Detergent Sanitizing Action of Disinfectants test method is used to validate the efficacy of food contact surface disinfectant/sanitizer with Gram-negative and Gram-positive bacterium. AOAC Germicidal Spray Product Test: 961.02 (Germicidal spray products as Disinfectants) is used to evaluate the efficacy of disinfectant with the spray method on hard, non-porous surfaces. It is a semi-quantitative method based on statistics of passing and far away from real life usage (extreme excess disinfectant quantity per unit surface area). Yet, this method has been modified and used for efficacy assessment of pre-saturated disinfecting towelettes.
ASTM (American Society for Testing and Materials) International is another important standards organization that develops the disinfectant efficacy tests. There are several methods published by ASTM for the effectiveness assessment in terms of different application strategy (liquid, wipes), application area (e.g. food contact surfaces or environmental surfaces and non-porous or porous surfaces etc.) and target microorganism (bacteria, fungi, mycobacteria, spores, biofilm, virus). They are mainly suspension or carrier-based test methods. What highlights in ASTM standards is it embraces several efficacy tests of pre-impregnated towelettes. For instance, ASTM E2362–15, a qualitative method (provide no quantitative reductions) by estimation of growth positive and negative to determine the effectiveness of pre-saturated or impregnated towelettes for hard surface disinfection. The listed materials (apparatus) for testing are easily accessible in a regular microbiology lab. It as well includes a large spectrum of testing organisms including mycobacteria. Similar to ASTM E2362–15 is the ASTM E2896–12, which determining the effectiveness of antimicrobial towelettes with a quantitative Petri plate method instead of a glass slide. Originally designed by Williams et al. in their three-steps protocol to determine the efficacy of disinfectant wipes on surfaces and later amended into standard ASTM E2967–15. In this standard, extra equipment named Wiperator is requested.
In above-mentioned published standards by different world recognised organisations, efficacy tests regarding wipe/towelette are very few and recent. These available quantitative test protocols are critically discussed below.
The EN 16615:2015 is a quantitative test method for the evaluation of bactericidal and yeasticidal activity on non-porous surfaces with mechanical action employing wipes for the medical area. This method displays several advantages allowing the quantitative evaluation of the antimicrobial activity of disinfecting wipes. It is applied to simulate the practical use of disinfecting wipes and allows to detect the cross-contamination caused by the wiping activity. Moreover, it can be used to evaluate the compatibility between the active ingredients in the solution and the wipe materials. It allows a flexible contact time (from 1 to 60 mins), can be tested in both clean and dirty conditions and define the declaration of concentration and exposure time on the disinfectants’ labels. Despite these advantages, this method also displays some drawbacks. The test is considered relatively time consuming, complex and is not possible to strictly control the applied mechanical action. Monotonic test surface (PVC with PUR surface coating) and the test wipe (if not specified by request), as well as the fixed disinfectant volume (16 ml), could significantly influence the outcome. Moreover, it is difficult to discriminate between the microbicidal activity derived from the disinfectant action (which represents the material compatibility issue) and the substrate that could retain microorganism by mere mechanical action.
The Modified AOAC international method 961.02 is meant for the disinfection evaluation performance of pre-saturated towelettes for hard surfaces. It is a simple method to study the variables that could influence the disinfection outcome. Approved by EPA as a method for the registration of spray disinfectants, this method gives a straightforward picture of test products’ performance providing survivor results in the form of a qualitative endpoint (growth positive versus growth negative). However, it exhibits unrealistic results when applied with a large ratio between disinfectant quantity and surface area. Besides, wiping applied pressure cannot be controlled, the concentration of bacteria on the test surface is not standardised, only allows semi-quantitative analysis, can only be applied in a monotonic surface (glass) and it is not possible to evaluate possible cross-contamination. Finally, because it does not address the humidity levels during the drying process of the test surfaces the results can be significantly uncertain.
The ASTM E2896–12 is a quantitative standard test method meant for the determination of the effectiveness of antimicrobial towelettes. The listed materials (apparatus) for testing the wipes are easily accessible in a regular microbiology lab and it requires easy operation procedures to evaluate the disinfecting-wipe ability using glass Petri dishes and corkscrew pattern wiping movements. Simple modifications can be done to test other microbial strain. Also, this method presents some disadvantages such as the lack of control of several variables of the disinfectant-impregnated wipe (disinfectant amount, wipe size, etc.), its exclusive use in monotonic surfaces (e.g. glass Petri dishes), the impossibility to evaluate cross-contamination and the uncontrolled wiping action, especially the wiping pressure. Another important drawback is the inability to differentiating between mechanical removal of inoculum from a surface and chemical inactivation of the test microbe.
The ASTM E2967–15 is a standard test method for assessing the ability of pre-wetted towelettes to remove and transfer bacterial contamination on hard, non-porous environmental surfaces using a specially designed machine to simulate the wiping action, the Wiperator. It allows great precision and reproducibility due to the well-control of the wiping action using the Wiperator. Despite this test is not widely recognised in Europe, it fills some gaps existing in other evaluation methods allowing a realistic contact time and a quantitative evaluation of the antimicrobial activity of disinfecting wipe. Moreover, it guarantees the evaluation of the disinfecting wipe’s ability to remove and prevent the microbial transfer from surfaces and their overall antimicrobial activity. However, also, in this case, the use of a monotonic test surface (stainless steel) and limited contact time (from 5 s up to 45 s) can limit a realistic outcome. There are some critics and debates related to the need for specific extra equipment (Wiperator) and whether the Wiperator could represent a realistic wiping process.
It is clear that an internationally recognized method to guarantees the evaluation of disinfecting wipe’s ability using quasi-realistic conditions, especially regarding test surfaces and cross-contamination, is urgently needed. Standards listed above are able to evaluate the overall antimicrobial efficacy of the testing wipes, but not differentiate the mechanical removal of inoculum from a surface and chemical inactivation of the test organisms. In addition, the wiping strategy should be addressed as one factor that can have an impact on the disinfection efficacy of DIWs. Divergent outcomes with different test standards can be suspected. A guideline for comparable results between various test standards is in demand.
Efficacy studies in literature
A countable number of studies regarding the efficacy of DIWs have been carried out. Tebbutt et al. may be the first ones, in 1988, to compare the decontamination performance of disposable and reusable disinfectant wipes concluding that the use of disposable disinfectant wipes significantly reduces the risks of cross-contamination. Moreover, their investigation was a breakthrough as it introduced the microbiological assessment of disinfecting wipes efficacy in practical use. The study, not only examined whether the wipe transferred bacteria from one surface to another but also if any organisms remaining on the wipe has been killed . Later, in 1993, Threlkeld et al. compared the disinfecting wipe method with the disinfectant soaking method in their efficacy to eliminate adenovirus 8 from medical instruments. The result revealed that the disinfectant wipe method could readily and thoroughly wipe away the virus from a tonometer and it was more convenient than disinfectant soaking method . However, their finding cannot be safely extrapolated to other equipment items, which implies different target surfaces and organic load that may have an impact on the decontamination performance of disinfecting wipes. In 2007, Williams et al. developed a three-step protocol to quantify the efficacy of disinfectant wipes, their ability to remove and prevent the microbial transfer from surfaces and their overall antimicrobial activity, which could be considered as a milestone for the development of efficacy test for disinfecting wipes . The paper introduced the first stringent test able to assess the ability of antimicrobial wipes to remove, kill and prevent the transfer of bacteria from contaminated surfaces. However, only one wiping material was used in this study, therefore no information can be extrapolated to understand the influence of different wiping materials in the surface disinfection efficacy using wipes. Afterwards, numerable studies have demonstrated the efficacy of disinfecting wipes based on the three-step protocol proposed by William et al. . Siani et al. (2011) tested 9 commercially available wipes from different manufacturers. Their study revealed the importance of application time in the sporicidal activity of disinfecting wipe . However, they did not investigate the role of wiping materials in conjunction with surface disinfectants. One innovation of their study is that they found spore binding to the wipe fibres, which gives more clues about the role of wiping materials in the disinfection process. In addition, the authors introduced the strategy “one wipe, one application, one direction”. Findings from Cadnum et al. gave a clear image of the efficient transfer of C. difficile spores from contaminated to clean surfaces using non-sporicidal wipes and the consistent reducing of C. difficile spores to undetectable levels at the inoculum site, with no transfer of spores to clean sites, using pre-moistened germicidal wipes . However, the active ingredient of non-sporicidal wipes was not reported.
After several studies with the Williams’ three-step protocol, it has been converted to the ASTM Standard E2967–15. The same year, Sattar et al. (2015) have published a paper regarding the efficacy of bioburden control from surfaces following disinfectant wipes use based on the new ASTM standard E2967–15. Five commercially available wipes have been tested with Staphylococcus aureus (ATCC 6538) and Acinetobacter baumannii (ATCC 19568) and their performance have been compared . One advance of this research is the newly added drying process, which eliminates the detrimental influence of it on microbial viability. Again, the combination of wiping materials and active ingredients is randomly reported, therefore the study of the interaction issue remains vague. Hernandez et al. (2008) have studied the disinfection performance of chlorine dioxide imbibed wipes against Mycobacterium avium based on the European standard prEN 14563 carrier test . However, their study was mainly focusing on mechanical action in the use of disinfecting wipes. There is more than one test method to assess the decontamination efficacy of disinfecting wipes. Gold et al. (2013) in their study have measured the cleanness, bacterial removal, and the force to remove the dried debris. Six commercially available disinfectant wipes were tested . The innovation part of their research is that they also evaluate the force and time required by the disinfectant cleaning wipes to remove the debris from the surface. However, the measurement methods (OPA assay and ATP bioluminescence assay) they adopted seem not to be very accurate. Despite this study gives hints for the selection of the disinfectant cleaning wipes, it is a case study with difficulties to apply for general use. The case study in MRSA-positive hospitalized patients from Cheng et al. (2011) has evaluated the effectiveness of disinfection with wipes against methicillin-resistant Staphylococcus aureus (MRSA) . Unfortunately, their experiment design showed a critical drawback. The post-disinfection swab only contained sterile saline solution instead of a neutralizer to counteract the sporicidal action from the disinfectant agent after one prescribed contact time.
The impact of pathogen transfer from fomites to fingers, using surface disinfecting wipes, has been evaluated by Lopez et al. (2014) in their research. Their study tested three different surfaces with four types of microorganisms, E. coli, S. aureus, B. thuringiensis, and PV-1. Their study has found that some microorganism may be more resistant to physical removal than others , which gives the clue that the adhesion of microorganism on wipes may be different depending on the type of material used.
The impact of the interaction between wiping material and surface disinfectant on the decontamination efficacy of disinfecting wipes was finally taken into account in the work of Engelbrecht et al. (2013). They have tested both cotton and microfiber towels on their abilities to bind quats using three different contact times. The study result indicated the reduction of quats concentration when exposed to cotton fibres, causing the disinfectant to fail the AOAC 961.02 Germicidal spray tests (GSTs) . Unfortunately, the microbiology tests they performed did not test the disinfecting wipe in their field use, because the AOAC 961.02 GST test does not consider the wipe in function of the microorganism removal during the wiping process. Thus, their study proved the deactivation of quats when exposing to cotton towels, but not the decontamination performance of quats disinfecting cotton towels. A list of the most important disinfecting wipes decontamination efficacy tests is summarised in Table 3.
Conclusion and future research
The use of pre-impregnated disinfecting wipes is one of the most efficient and prevalent methods for the decontamination of high-touch environmental surfaces and non-critical medical devices in hospitals and other healthcare centres. There is evidence to support the significance of disinfecting wipes in preventing cross-contamination and spread of HCAIs. Despite this, less is known concerning the effectiveness of disinfecting wipes in the decontamination process. From the studies, several variables influence the disinfection efficacy of DIWs besides the external factors, these include:
Disinfectant (type, concentration)
Wipe (material, construction)
Wiping strategy including the applied pressure force, wiped surface area, the geometry of the mechanical action, number of passages, and remaining time on the surface.
Storage time (function degradation)
Amongst, the interaction between disinfectant and textile substrate is the biggest encumbrance for its disinfection performance. Though, literature has revealed that an inappropriate material of the wipes could interact with the adsorbed active ingredient resulting in lower or even abolished disinfectant efficacy. At present, there is no clear understanding of the interaction phenomenon. Several information gaps have to be filled to obtain consistent and exhaustive knowledge about the interaction, in particular, examining and improving the following issues:
Material compatibility (combination of wipe and disinfectant)
Liquor ratio (wipe mass/disinfection solution volume)
Contact time (of disinfectant and wipes)
Besides, the standards to date remain some drawbacks in testing the effectiveness of DIWs. For example, difficulties exist in differentiating between mechanical removal of inoculum from a surface and chemical inactivation of the test microbe (High risk of cross-contamination when pathogens are just being removed by the wipe instead of being killed by the disinfectant depending on the materials compatibility). More realistic disinfectant volume per unit surface area needs to be improved and applied. Divergent outcomes with different test standards can be suspected. A guideline for comparable results between various test standards is in demand. Nowadays, the most reliable method that can be used in hospitals seems to be the one using ready-to-use disinfecting wipes because of its lower disinfection failure risk. Due to the incomplete study of the decontamination efficacy of DIWs and the lack of testing standards validating the efficacy of DIWs in nosocomial practice, one can hardly advocate for their use in hospitals.
Therefore, one future research direction could focus on the interaction mechanism and its impact on the DIWs’ overall decontamination activity. Furthermore, emphasis should be placed on the ageing of DIWs over storage time in respects to the structure or properties deterioration of the wiping material as well as the antimicrobial efficacy change. Moreover, almost nothing is known about properties, performance and disinfection efficiency of plasma-treated and/or polymer-functionalized wipes. Efficacy test with advanced surface modification technologies can also be considered for future research direction. The development of more environmentally sustained processes is also required considering the waste management of disposable wipes. Additionally, the study on the effectiveness of disinfectant pre-impregnated wipes using the appropriate materials will avoid wasting resources. The outcome research knowledge will be important to ensure hospitals daily workflow from unnecessary risk of infection outbreak and to complement the products’ user manual of disinfectant and wipes in the market.
Availability of data and materials
Accelerated hydrogen peroxide
Association of Official Analytical Chemists
American Society for Testing Materials
Commercially available wipe product
American Centres for Disease Control and Prevention
European Committee for Normalization
European Chemicals Agency
Environmental Protection Agency
Germicidal spray tests
Healthcare Infection Control Practices Advisory Committee
Methicillin-resistant Staphylococcus aureus
National Center for Biotechnology Information
Office of Pesticide Programs
Office of Prevention, Pesticides & Toxic Substances
Pre-impregnated disinfecting wipe (pre-wetted disinfecting wipe)
Pre-soaked disinfecting wipe (bucket method)
Quaternary Ammonium Compounds (quats)
Quantitative microbial risk assessment
“Ready-to-use” disinfecting wipe
- TC 216:
Technical Committee 216
Kassis C, Hachem R, Raad II, Perego CA, Dvorak T, Hulten KG, et al. Outbreak of community-acquired methicillin-resistant Staphylococcus aureus skin infections among health care workers in a cancer center. Am J Infect Control. 2011;39:112–7.
Gold KM, Hitchins VM. Cleaning assessment of disinfectant cleaning wipes on an external surface of a medical device contaminated with artificial blood or Streptococcus pneumoniae. Am J Infect Control. 2013;41:901–7.
Albrich WC, Harbarth S. Health-care workers: source, vector, or victim of MRSA? Lancet Infect Dis. 2008;8:289–301.
Klevens RM, Edwards JR, Richards CL, Horan TC, Gaynes RP, Pollock DA, et al. Estimating health care-associated infections and deaths in U.S. hospitals, 2002. Public Health Rep. 2016;122:160–6.
Otter JA, Yezli S, French GL. The role played by contaminated surfaces in the transmission of nosocomial pathogens. Infect Control Hosp Epidemiol. 2015;32:687–99.
Kramer A, Schwebke I, Kampf G. How long do nosocomial pathogens persist on inanimate surfaces? A systematic review. BMC Infect Dis. 2006;6:130.
Yuen J, Chung T, Loke A. Methicillin-resistant Staphylococcus aureus (MRSA) contamination in bedside surfaces of a hospital Ward and the potential effectiveness of enhanced disinfection with an antimicrobial polymer surfactant. Int J Environ Res Public Health. 2015;12:3026–41.
Otter JA, Yezli S, French GL. The role of contaminated surfaces in the transmission of nosocomial pathogens. Use of Biocidal Surfaces for Reduction of Healthcare Acquired Infections; 2014. p. 27–58.
Weber DJ, Anderson D, Rutala WA. The role of the surface environment in healthcare-associated infections. Curr Opin Infect Dis. 2013;26:338–44.
Otter JA, Yezli S, Salkeld JAG, French GL. Evidence that contaminated surfaces contribute to the transmission of hospital pathogens and an overview of strategies to address contaminated surfaces in hospital settings. Am J Infect Control. 2013;41:S6–S11.
Weber DJ, Rutala WA, Miller MB, Huslage K, Sickbert-Bennett E. Role of hospital surfaces in the transmission of emerging health care-associated pathogens: norovirus, Clostridium difficile, and Acinetobacter species. Am J Infect Control. 2010;38:S25–33.
Giraudon I, Cathcart S, Blomqvist S, Littleton A, Surman-Lee S, Mifsud A, et al. Large outbreak of salmonella phage type 1 infection with high infection rate and severe illness associated with fast food premises. Public Health. 2009;123:444–7.
Podolak R, Enache E, Stone W, Black DG, Elliott PH. Sources and risk factors for contamination, survival, persistence, and heat resistance of Salmonella in low-moisture foods. J Food Prot. 2010;73:1919–36.
Donskey CJ. Does improving surface cleaning and disinfection reduce health care-associated infections? Am J Infect Control. 2013;41:S12–S9.
Rutala WA, Weber DJ. Disinfectants used for environmental disinfection and new room decontamination technology. Am J Infect Control. 2013;41:S36–41.
Dancer SJ. Controlling hospital-acquired infection: focus on the role of the environment and new Technologies for Decontamination. Clin Microbiol Rev. 2014;27:665–90.
Weinstein RA, Hota B. Contamination, disinfection, and cross-colonization: are hospital surfaces reservoirs for nosocomial infection? Clin Infect Dis. 2004;39:1182–9.
Rutala WA, White MS, Gergen MF, Weber DJ. Bacterial contamination of keyboards: efficacy and functional impact of disinfectants. Infect Control Hosp Epidemiol. 2016;27:372–7.
Eckstein BC, Adams DA, Eckstein EC, Rao A, Sethi AK, Yadavalli GK, et al. Reduction of clostridium difficile and vancomycin-resistant Enterococcus contamination of environmental surfaces after an intervention to improve cleaning methods. BMC Infect Dis. 2007;7:61.
Deshpande A, Mana TSC, Cadnum JL, Jencson AC, Sitzlar B, Fertelli D, et al. Evaluation of a sporicidal Peracetic acid/hydrogen peroxide–based daily disinfectant cleaner. Infect Control Hosp Epidemiol. 2016;35:1414–6.
Amy G, Bull R, Craun GF, Pegram R, Siddiqui M. Organization WH Disinfectants and disinfectant by-products; 2000.
Boyce JM. Modern technologies for improving cleaning and disinfection of environmental surfaces in hospitals. Antimicrob Resist Infect Control. 2016;5:1–10.
Kundrapu S, Sunkesula V, Jury LA, Sitzlar BM, Donskey CJ. Daily disinfection of high-touch surfaces in isolation rooms to reduce contamination of healthcare Workers' hands. Infect Control Hosp Epidemiol. 2015;33:1039–42.
Panousi MN, Williams GJ, Girdlestone S, Hiom SJ, Maillard JY. Evaluation of alcohol wipes used during aseptic manufacturing. Lett Appl Microbiol. 2009;48:648–51.
Hernández A, Carrasco M, Ausina V. Mycobactericidal activity of chlorine dioxide wipes in a modified prEN 14563 test. J Hosp Infect. 2008;69:384–8.
Rengasamy RS. Composite nonwovens in wipes. Composite Non-Woven Materials; 2014. p. 89–119.
Soukupova V, Boguslavsky L, AnandJiwala RD. Studies on the properties of biodegradable wipes made by the Hydroentanglement bonding technique. Text Res J. 2016;77:301–11.
Edwards NWM, Best EL, Connell SD, Goswami P, Carr CM, Wilcox MH, et al. Role of surface energy and nano-roughness in the removal efficiency of bacterial contamination by nonwoven wipes from frequently touched surfaces. Sci Technol Adv Mater. 2017;18:197–209.
Williams GJ, Denyer SP, Hosein IK, Hill DW, Maillard JY. The development of a new three-step protocol to determine the efficacy of disinfectant wipes on surfaces contaminated with Staphylococcus aureus. J Hosp Infect. 2007;67:329–35.
Moore G, Griffith C. A laboratory evaluation of the decontamination properties of microfibre cloths. J Hosp Infect. 2006;64:379–85.
Gillespie E, Brown R, Treagus D, James A, Jackson C. Improving operating room cleaning results with microfiber and steam technology. Am J Infect Control. 2016;44:120–2.
Abernethy M, Gillespie E, Snook K, Stuart RL. Microfiber and steam for environmental cleaning during an outbreak. Am J Infect Control. 2013;41:1134–5.
Bergen LK, Meyer M, Høg M, Rubenhagen B, Andersen LP. Spread of bacteria on surfaces when cleaning with microfibre cloths. J Hosp Infect. 2009;71:132–7.
Suh H, Duckett K, Bhat G. Biodegradable and tensile properties of cotton/cellulose acetate nonwovens. Text Res J. 2016;66:230–7.
Edana I. In: Edana I, editor. Guidelines for assessing the Flushability of disposable nonwoven products. Brussels: European Disposables and Nonwovens Association, Association of the Nonwoven Fabrics Industry; 2013.
Maris P. Modes of action of disinfectants. Rev Sci Tech. 1995;14:47.
Abreu AC, Tavares RR, Borges A, Mergulhao F, Simoes M. Current and emergent strategies for disinfection of hospital environments. J Antimicrob Chemother. 2013;68:2718–32.
Suchomel M, Rotter M. Ethanol in pre-surgical hand rubs: concentration and duration of application for achieving European norm EN 12791. J Hosp Infect. 2011;77:263–6.
Eggerstedt S. Comparative efficacy of commercially available alcohol-based hand rubs and World Health Organization-recommended hand rubs. Am J Infect Control. 2013;41:472–4.
Threlkeld AB, Froggatt JW, Schein OD, Forman MS. Efficacy of a disinfectant wipe method for the removal of adenovirus 8 from tonometer tips. Ophthalmology. 1993;100:1841–5.
Chang SC, Li WC, Huang KY, Huang YC, Chiu CH, Chen CJ, et al. Efficacy of alcohols and alcohol-based hand disinfectants against human enterovirus 71. J Hosp Infect. 2013;83:288–93.
Huerkamp MJ. Alcohol as a disinfectant for aseptic surgery of rodents: crossing the thin blue line? J Am Assoc Lab Anim Sci. 2002;41:10–2.
Coates D. An evaluation of the use of chlorine dioxide (Tristel one-shot) in an automated washer/disinfector (Medivator) fitted with a chlorine dioxide generator for decontamination of flexible endoscopes. J Hosp Infect. 2001;48:55–65.
Weber DJ, Consoli SA, Rutala WA. Occupational health risks associated with the use of germicides in health care. Am J Infect Control. 2016;44:e85–e9.
Kujala VM, Reijula KE, Ruotsalainen EM, Heikkinen K. Occupational asthma due to chloramine-T solution. Respir Med. 1995;89:693–5.
Wenger M. Join the A-team: exploring the key role nurses play in antimicrobial stewardship. Can J Crit Care Nurs. 2017;28:50–50.
Wessels S, Ingmer H. Modes of action of three disinfectant active substances: a review. Regul Toxicol Pharmacol. 2013;67:456–67.
Russell A. Chemical sporicidal and sporostatic agents. In: Block SS editor. Disinfection, Sterilization, and Preservation. Philadelphia: Lippincott Williams & Wilkins; 2001. p. 529–541.
Bounoure F. Comparison of hydrogen peroxide and peracetic acid as isolator sterilization agents in a hospital pharmacy. Am J Health Syst Pharm. 2006;63:451–5.
Omidbakhsh N, Sattar SA. Broad-spectrum microbicidal activity, toxicologic assessment, and materials compatibility of a new generation of accelerated hydrogen peroxide-based environmental surface disinfectant. Am J Infect Control. 2006;34:251–7.
Heseltine P. In: Block SS, editor. Disinfection, Sterilization, and Preservation. 5th ed. Philadelphia: Lippincott Williams & Wilkins; 2001. 1,504 pages. Infect Control Hosp Epidemiol 2015; 23: 109.
Hinchliffe DJ, Condon BD, Slopek RP, Reynolds M. The adsorption of alkyl-dimethyl-benzyl-ammonium chloride onto cotton nonwoven hydroentangled substrates at the solid–liquid interface is minimized by additive chemistries. Text Res J. 2016;87:70–80.
Tezel U, Pavlostathis SG. Role of quaternary ammonium compounds on antimicrobial resistance in the environment. Antimicrobial Resistance in the Environment; 2011. p. 349–87.
Gerba CP, Müller V. Quaternary ammonium biocides: efficacy in application. Appl Environ Microbiol. 2015;81:464–9.
Jennings MC, Minbiole KPC, Wuest WM. Quaternary ammonium compounds: an antimicrobial mainstay and platform for innovation to address bacterial resistance. ACS Infect Dis. 2015;1:288–303.
Møretrø T, Heir E, Nesse LL, Vestby LK, Langsrud S. Control of Salmonella in food related environments by chemical disinfection. Food Res Int. 2012;45:532–44.
Ghanbar S, Kazemian MR, Liu S. New generation of N-chloramine/QAC composite biocides: efficient antimicrobial agents to target antibiotic-resistant Bacteria in the presence of organic load. ACS Omega. 2018;3:9699–709.
Jiao Y, L-n N, Ma S, Li J, Tay FR, J-h C. Quaternary ammonium-based biomedical materials: state-of-the-art, toxicological aspects and antimicrobial resistance. Prog Polym Sci. 2017;71:53–90.
Ecolab. Quat Absorption Onto Textiles ECOLAB; 2010. p. 8.
Spiller HA, Quadrani-Kushner DA, Cleveland PA. Five year evaluation of acute exposures to phenol disinfectant (26%). J Toxicol Clin Toxicol. 2008;31:307–13.
Moorer WR. Antiviral activity of alcohol for surface disinfection. Int J Dent Hyg. 2003;1:138–42.
Kusumaningrum HD, Paltinaite R, Koomen AJ, Hazeleger WC, Rombouts FM, Beumer RR. Tolerance of Salmonella Enteritidis and Staphylococcus aureus to surface cleaning and household bleach. J Food Prot. 2003;66:2289–95.
Williams GJ, Denyer SP, Hosein IK, Hill DW, Maillard J-Y. Limitations of the efficacy of surface disinfection in the healthcare setting. Infect Control Hosp Epidemiol. 2015;30:570–3.
Tebbutt GM. Laboratory evaluation of disposable and reusable disinfectant cloths for cleaning food contact surfaces. Epidemiol Infect. 2009;101:367–375.
Cheng KL, Boost MV, Chung JWY. Study on the effectiveness of disinfection with wipes against methicillin-resistant Staphylococcus aureus and implications for hospital hygiene. Am J Infect Control. 2011;39:577–80.
Qutaishat S, Teska P. Quantitative Analysis of Materials and Methods in Cleaning and Disinfection of Environmental Surfaces: Microfiber vs. Cotton and Spray vs. Soak. Am J Infect Control. 2012;40:e40.
Han JH, Sullivan N, Leas BF, Pegues DA, Kaczmarek JL. Umscheid CA cleaning hospital room surfaces to prevent health care–associated infections. Ann Intern Med. 2015;163:598.
Boyce JM, Sullivan L, Booker A, Baker J. Quaternary ammonium disinfectant issues encountered in an environmental services department. Infect Control Hosp Epidemiol. 2015;37:340–2.
Bloß R, Meyer S, Kampf G. Adsorption of active ingredients of surface disinfectants depends on the type of fabric used for surface treatment. J Hosp Infect. 2010;75:56–61.
Alfa MJ, Lo E, Olson N, MacRae M, Buelow-Smith L. Use of a daily disinfectant cleaner instead of a daily cleaner reduced hospital-acquired infection rates. Am J Infect Control. 2015;43:141–6.
Wiemken TL, Curran DR, Pacholski EB, Kelley RR, Abdelfattah RR, Carrico RM, et al. The value of ready-to-use disinfectant wipes: compliance, employee time, and costs. Am J Infect Control. 2014;42:329–30.
Masuku SM, Babu D, Martin EM, Koo OK, O’Bryan CA, Crandall PG, et al. Cleaning and decontamination efficacy of wiping cloths and silver dihydrogen citrate on food contact surfaces. J Appl Microbiol. 2012;113:89–95.
Cadnum JL, Hurless KN, Kundrapu S, Donskey CJ. Transfer of Clostridium difficile spores by nonsporicidal wipes and improperly used hypochlorite wipes practice + product = perfection. Infect Control Hosp Epidemiol. 2015;34:441–2.
West AM, Nkemngong CA, Voorn MG, Wu T, Li X, Teska PJ, et al. Surface area wiped, product type, and target strain impact bactericidal efficacy of ready-to-use disinfectant Towelettes. Antimicrob Resist Infect Control. 2018;7:1–8.
Nam S, Slopek R, Condon BD, Sawhney P. Effect of polyester blends in hydroentangled raw and bleached cotton nonwoven fabrics on the adsorption of alkyl-dimethyl-benzyl-ammonium chloride. Text Res J. 2014;85:1221–33.
Hinchliffe DJ, Condon BD, Madison CA, Reynolds M, Hron RJ. An optimized co-formulation minimized quaternary ammonium compounds adsorption onto raw cotton disposable disinfecting wipes and maintained efficacy against methicillin-resistant Staphylococcus aureus, vancomycin-resistant Enterococcus faecalis, and Pseudomonas aeruginosa. Text Res J. 2017;88:2329–38.
Powell WA, Binford L, Mattingly W, Kelley R, Guinn B, Peyrani P, et al. Disinfectant sprays or wipes: applications in the setting of behavioral health. Am J Infect Control. 2015;43:S20.
Del Re D, Ikeno C, Smid K, Swift D. Effects of disinfectant wipes on touch screen surfaces. Am J Infect Control. 2015;43:S20–S1.
Merritt M, Brown R, Glenn A, Chudasama Y. Eberhardt M 255 effect of antimicrobial disinfectant wipes on Bacteria on computer equipment in the emergency department. Ann Emerg Med. 2015;66:S92.
van Klingeren B. A brief history of European harmonization of disinfectant testing - a Dutch view. GMS Krankenhhyg Interdiszip. 2007;2:Doc14.
Agency USEP. Production performance test guidelines OCSPP 8102200: Disinfectants for Use on Hard Surface - Efficacy Data Recommendations: Office of Chemical Safety and Pollution Prevention (7510P), EPA, vol. 29; 2012.
Siani H, Cooper C, Maillard J-Y. Efficacy of “sporicidal” wipes against Clostridium difficile. Am J Infect Control. 2011;39:212–8.
Sattar SA, Bradley C, Kibbee R, Wesgate R, Wilkinson MAC, Sharpe T, et al. Disinfectant wipes are appropriate to control microbial bioburden from surfaces: use of a new ASTM standard test protocol to demonstrate efficacy. J Hosp Infect. 2015;91:319–25.
Lopez GU, Kitajima M, Havas A, Gerba CP, Reynolds KA, Schaffner DW. Evaluation of a disinfectant wipe intervention on fomite-to-finger microbial transfer. Appl Environ Microbiol. 2014;80:3113–8.
Engelbrecht K, Ambrose D, Sifuentes L, Gerba C, Weart I, Koenig D. Decreased activity of commercially available disinfectants containing quaternary ammonium compounds when exposed to cotton towels. Am J Infect Control. 2013;41:908–11.
Lopez GU, Kitajima M, Sherchan SP, Sexton JD, Sifuentes LY, Gerba CP, et al. Impact of disinfectant wipes on the risk ofCampylobacter jejuniinfection during raw chicken preparation in domestic kitchens. J Appl Microbiol. 2015;119:245–52.
Xinyu Song (SFRH/BD/130028/2017) acknowledges Fundação para a Ciência e Tecnologia (FCT), Portugal, for its doctoral grant financial support. Andrea Zille also acknowledges financial support of the FCT through an Investigator FCT Research contract (IF/00071/2015) and the project POCI-01-0145-FEDER-007136 and UID/CTM/00264 under the COMPETE and FCT/MCTES (PIDDAC) co-financed by European funds (FEDER) through the PT2020 program.
This work was supported by the Fundação para a Ciência e Tecnologia (FCT) [grant number: SFRH/BD/130028/2017].
Ethics approval and consent to participate
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.